This item ships for FREE with 2-Day Shipping on orders over $50. (Details)
Want it delivered on Wednesday, December 11? Order in-stock items in the next 17 hours 12 minutes and choose One-Day Shipping at checkout.
Package Size: 19.1 x 15.7 x 15 inches
This Thermo Scientific ELED Model Safe-T SH9 Thermolyne explosion-proof stirring hot plate requires 120V, and is suitable for use in hazardous environments where flammable gases or solvent vapors can produce explosive or ignitable mixtures. Stirring hot plate features electronic temperature control that maintains temperature stability to ±4degF (±2.0degC). An automatic overtemperature safety switch cuts off power if the temperature exceeds 536degF (280degC). Cast-aluminum top plate provides efficient heat transfer and excellent temperature uniformity. The powder-coat painted, stainless steel housing is corrosion-resistant and easy to clean. Includes a 1/2 inch conduit attachment, and an 18 inch lead wire for UL require. Optional dial thermometer, with 0 Degree to 300 Degree C range, fits directly into top plate and measures surface temperature. Large, aluminum top plate surface measures 9.1 x 9.1in (23.2 x 23.2cm) and provides efficient heat transfer. Precise electronic heat control maintains temperature stability within ±2 Degree C (±4 Degree F). Electronic temperature controls range from 38 to 220 Degree C (100 to 428 Degree F). Stir solutions with a viscosity of 1200cp at 400rpm (at 21.5 Degree C). Sturdy construction allows maximum load capacity, 25lb. (11.3kg) on top plate. Stainless-steel case is easily cleaned and maintained, and accommodates a metal vessel. Heating Surface: 9.1 x 9.1in. Temperature Range: 38 to 220 Degree C (100 to 428 Degree F). Stirring Speed Range: 60 to 1200rpm. Overall Dimensions (L x W x H): 12.8 x 10 x 8.1in. Voltage Rating: 120V. Frequency: 50/60Hz. Power Rating: 1070 watts. Current Rating: 8.9A. Warranty: Three Years. Certifications: UL/cUL Listed. Thermo Scientific ELED #: SP87325. Thermo Scientific ELED - "Every Lab, Every Day".
How can we improve our Product Description?
There are no customer reviews yet.Be the first to review this item