This item ships for FREE with 2-Day Shipping on orders over $50. (Details)
Want it delivered on Wednesday, May 29? Order in-stock items in the next 3 days 19 hours 38 minutes and choose One-Day Shipping at checkout.
Package Size: 15.3 x 12.3 x 4.5 inches
The HI 96733C is an auto-diagnostic, portable microprocessor meter that measures the ammonia nitrogen content in water samples in the 0.0 to 50.0 mg/L (ppm) range. Hanna’s CAL CHECK function validates instrument performance. The meter is splash waterproof and the LED and filter units are protected from dust or dirt by a transparent cup, making it suitable for field applications. Messages on the LCD screen aid the user in routine operation and an auto-shut off feature turns the meter off automatically after 10 minutes of non use in measurement mode or after one hour if left in calibration mode. The device uses an exclusive positive-locking system to ensure cuvette is in the same place every time it is placed into the measurement cell and fits a cuvette with a larger neck, making it easier to add both sample and reagents.
Range is 0.0 to 50.0 mg/L (as ammonia-nitrogen) + or – 0.5 mg/L + or – 5% of reading at 25 degrees C with a resolution of 0.1 mg/L for measurement, calibration, and validation. The device operates on a 9V battery. Dimensions are 192 x 104 x 69 mm (7.6” x 4.1” x 2.7’’), and weighs 360 g (12.7 oz).
Ammonia is often an indication of the presence of animal or plant microbiological decay. Ammonia photometers are used by researchers, manufacturers, and individuals to measure the ammonia-nitrogen content in lakes, rivers, portable water, builder feed water, sewage, industrial, and wastewater.
Hanna Instruments manufactures analytical instrumentation and supplies, including pH meters, electrodes, chemical reagents, and buffer solutions. The company, founded in 1978 and headquartered in Woonsocket, RI, meets International Organization for Standardization (ISO) 9001 standards.
What’s in the Box?
How can we improve our Product Description?
There are no customer reviews yet.Be the first to review this item