This item ships for FREE with 2-Day Shipping on orders over $50. (Details)
Want it delivered on Tuesday, May 28? Order in-stock items in the next 18 hours 54 minutes and choose One-Day Shipping at checkout.
Package Size: 13.1 x 10.3 x 4.6 inches
The HI 255 is a multi-parameter micro-processor-based pH, oxidation reduction potential (ORP), conductivity, salinity and Total Dissolved Solids (TDS) bench meter.
The meter features a large LCD screen, automatic temperature compensation (ATC) or manual compensation, and calibration at up to three points, with five memorized buffer solutions pH 4.01, 6.86, 7.01, 9.18, 10.01.
pH range is -2.00 to 16.00 (+ or - 0.01pH) with a resolution of 0.01 pH, or -2.000 to 16.000 (+ or - 0.002 pH) with a resolution of 0.001 pH. Temperature range is - 10.0 to 120.0 degrees C, ORP range is + or - 699.9 mV or 2000 mV, and salinity range is 0.0 to 400.0% NaCl. Conductivity range is 0.00 to 29.99 uS/cm; 30.0 to 299.9 uS/cm; 300 to 2999 uS/cm; 3.00 to 29.99 mS/cm; 30.0 to 200.0 mS/cm; up to 500.0 mS/cm actual conductivity. TDS range is 0.00 to 14.99 ppm; 15.0 to 149.9 ppm; 150 to 1499 ppm; 1.50 to 14.99 g/l; 15.0 to 100.0 g/l; up to 400.0 g/l actual TDS (with 0.80 factor). Dimensions are 240 by 182 by 74 mm (9.4 by 7.1 by 2.9”), with weight 1.1 kg (2.5 lbs); kit with holder is 2.5 kg (5.5 lbs).
pH testers measure the acidity or alkalinity of a substance and are commonly used by researchers, manufacturers, and hobbyists.
Hanna Instruments manufactures a wide variety of analytical instrumentation, including pH meters, multi-parameter meters, electrodes, chemical reagents and buffer solutions. Founded in 1978, Hanna is headquartered in Woonsocket, RI, with subsidiaries in 32 countries. Hanna Instruments holds many firsts in the field of analytical instrumentation, including the world’s first pH electrode with a built-in temperature sensor, electronic pocket-sized pH tester, and replaceable electrode pH pocket tester. To ensure the quality of its products, Hanna is a vertically integrated manufacturer and does not subcontract any part of its manufacturing.
What’s in the Box?
How can we improve our Product Description?
There are no customer reviews yet.Be the first to review this item