This item ships for FREE with 2-Day Shipping on orders over $50. (Details)
Want it delivered on Wednesday, May 29? Order in-stock items in the next 3 days 3 hours 36 minutes and choose One-Day Shipping at checkout.
Package Size: 11 x 10.9 x 9.4 inches
The controller/pump is enclosed in a modular housing for maximum protection. The pump utilizes a one-piece polypropylene, injection-molded housing. Since there are no joints or screws holding different sections of the housing together, the case is extremely rugged and sturdy. This series of instruments will mount easily in your plant using a minimum of wall space. The controls and pump head are located in the front to allow easy access. ORP controller and dosing pumps in one compact unit. +/-5mV accuracy with unbeatable performance. Isolated 4 to 20mA recorder output. Proportional dosing slows the pump down when the measured ORP level approaches the set value, to avoid over dosage of oxidizing or reducing agents. Alarm contact is activated whenever the ORP reading varies more than 200mV from the set point. Auxiliary contacts allow users to attach a mixer or priming pump that is activated only when the pump is dosing. PVDF, FPM/FKM and PTFE materials are used for all parts that come into contact with liquid. This model is supplied with discharge and suction valves, polyethylene tubing, 115V power cable and instructions. -999mV to +999mV Range. 1mV Resolution. Flow Rate: 13.3 LPH on 0.5 bars pressure, 11.7 LPH on 1.0 bars pressure, 10.1 LPH on 2.0 bars pressure, 9.0 LPH on 3.0 bars pressure, 7.8 LPH on 4.0 bars pressure. 10(12) Ohm input impedance. Proportional, acid or base, user selectable dosage. Dosing Contact; 1 isolated contact, 2A, Max. 240V, resistive load, 1,000,000 strokes. Alarm contact: 1 isolated contact, 2A, Max. 240V, resistive load, 1000000 strokes. Power supply: 115V +/-15 percent; 50/60Hz (40W). Environment: 0 to 50 degree C (32 to 122 degree F); RH max 95 percent non-condensing. Measures 8-11/16" length by 5-19/32" width by 7-3/32" height.
How can we improve our Product Description?
There are no customer reviews yet.Be the first to review this item