This item is no longer available
This item is NOT eligible for "Free Standard Shipping on orders over $50.00" promotion. (Details)
Package Size: 9.9 x 6.3 x 3.5 inches
The process meter puts a complete troubleshooting solution in the palm of your hand. Get in touch with more information, faster and easier. The ProcessMeter gives you the measurement capabilities you expect from our rugged, reliable DMMs plus the ability to source, simulate, or measure dc current with 1 microamp resolution and accuracy of 0.05 percent. Put it all together and you have the ideal tool for troubleshooting and calibrating current loop applications. So if you're tired of always tracking down a calibrator, lugging around two tools and switching back and forth, get the single best solution for doubling your power. 25 percent manual step plus auto step and auto ramp on ma output. Current measurement: range - 0-1 A/0-30 mA, resolution - 1 mA/0.001 mA, accuracy - 0.2 percent + 2 LSD/0.05 percent + 2 LSD. Current sourcing: range - 0-20 mA or 4-20 mA, resolution - 0.05 percent of span, Drive capability 500 ohm at 24 mA. Frequency: to 19.999 kHz, 0.005 percent + 1 LSD. Continuity: Beeps for resistance less-than 100 ohms. Voltage measurement: range - 0-1000 V AC/DC, resolution - 0.1 mV to 1.0 V, accuracy 0.1 percent Rdg + 1 LSD (VDC). Resistance measurement: to 40 MO, 0.2 percent + 1 LSD. Diode test: 2.4 V shows diode voltage drop. Operating temperature: -20 to 55 degree C. Storage temperature -40 to 60 degree C. Humidity (without condensation): 95 percent up to 30 degree C, 75 percent up to 40 degree C, 45 percent up to 50 degree C, 35 percent up to 55 degree C. Electrical safety: designed in accordance with IEC 1010-1, ANSI/ISA S82.01-1994 and CAN/CSA C22.2 No. 1010.1-92. Overvoltage category: IEC 1010-1 CAT III. Maximum voltage: 1000 V rms. Batteries: single 9V alkaline battery (ANSI/NEDA 1604A or IEC 6LR61). 3 years warranty. Battery life: 50 hours typical (measurement), 12 hours typical (sourcing 12 mA). 187mm length by 87mm width by 32mm height.
How can we improve our Product Description?
There are no customer reviews yet.Be the first to review this item